This is some intro text…
Contact info specific to the equipment, if different than the info collected for the client organization.
All tests need to pass
Sash shall be raised 1" above and/or below the recommended height and alarm.
Audible & visible alarm shall respond with 15 seconds after the exhaust fan has been reduced.
Total noise level shall not exceed 70 dbA when background noise < 60 dbA. If background noise is > 60 dbA then the level will be corrected per standard correction chart.
Lighting shall be no less than 45 FC greater than background levels where background levels average a maximum of 15 FC
Reason for contacting us:
---Laboratory Testing ServicesCustomer ServiceLab Relocation ServicesOther
Enter the destination URL
Or link to existing content